يتم توفير ذلك من خلال مراقبة وإضافة هيبوكلوريت الصوديوم (NaOCl) في ماء الغسل والسوائل ويمكن استخدامه كنقطة انطلاق لتطوير إجراءات تشغيلية خاصة بالمرافق الصحية.
Cl-1 اختبار هيبوكلوريت الصوديوم
Fast, straightforward results
Sodium hypochlorite (NaOCl) as the standard solution
Large LCD screen easy to read
Accuracy(NaClO solution): 0~ 50ppm: +-2%+1d , 50~150 ppm: +/-3%+1d , 150~1200 ppm: +/-5%+1d
Automatic Shutdown after 40 seconds of Non-Use
Requires 1 CR3032 3V Lithium Batteries
Size/Weight: 128*25*20mm, 35g
Determine your NaOCl level
Fresh fruit and vegetables packers: 50-150ppm
Non-porous food contact surface: 100-200ppm
Porous surfaces: 600ppm
Floors and Walls: 1000ppm
NaClO concentrations that are too high can damage the product and harm employees.It is wise to wear protective clothing and eye covering when using NaClO solutions. Also, make sure that the area is well ventilated. Harmful chlorine gas can be producted if the solution falls below pH 4.0 or if the NaClO is used in hot water.
It will take some time to determine how often you will need to check NaClO concentration and add more NaClO to your system. The first time you use the Chlorine meter, check concentration (ppm) levels frequently. If you find that the chlorine is being used up too quickly and you have lots of organic matter (dirts, residue etc.) in your wash water, consider changing the water more often. High levels of inorganic materials (such as iron) in water can alter ppm readings.
Automatic Temperature Compensation (ATC) self-correcting takes one minute to complete.
Probe close to container or surface, the mearsured value will be lower.